Top tips for... analyser maintenance
Keeping continuous water analysers well maintained is vital to ensure you can rely on the information they provide, writes Jon Farrington
by Jon Farrington, Technical Manager, ABB
Continuous water analysers are useful tool to help meet tighter environmental regulations and rising customer expectations, but their performance is only as good as the condition of the analyser itself.
With the introduction of ever more stringent standards by the UK Environment Agency, it is particularly important to maintain analysers to the highest possible levels as a matter of best practice. In the event of a water quality incident, operators must demonstrate that their monitoring and measurement equipment was being correctly maintained.
With analysers very much on the front line of every water treatment process, the need to ensure they are kept in good working order is critical to avoid any penalties arising from a failure. Following the steps outlined here will provide a useful starting point for helping to design an effective maintenance strategy and prolong the life of continuous water analysers.
1. Heed the warning signs
Many of the latest generation of analysers incorporate self-diagnostic features and alarm capabilities to warn when something is wrong. Although these signals are useful in determining when maintenance needs to be performed, it is nevertheless advisable to have a preventive maintenance regime in place to help extend the life of the instrument.
2. Check reagent levels
Reagents are crucial for producing the chemical reaction needed for accurate measurement to take place. As such, it is important to ensure that reagents are replenished as soon as they are exhausted. Checking reagent consumption regularly will help you predict the best time to carry out a changeover. As reagents have a shelf life, check the expiration date before exchanging the used reagent with the new one. In addition, regular maintenance will also help to show whether the reagent is being consumed at the rate that should be expected for the application. Under or over consumption can be indicative of a problem in the analyser’s fluid handling system, such as a blockage or restriction.
3. Perform regular visual checks
Analyser functions and readings should be thoroughly checked to ensure they work at optimum efficiency. Checking for leaks (pipe connections and tubing and liquid handling components), liquid levels (reagents, calibration and cleaning solution bottles) and potential obstructions to the sample flow should be performed as regularly as possible. It’s worth remembering that erratic readings on the analyser display unit may not be indicative of a fault with the transmitter, but rather external factors such as unexpected changes in water quality. Where doubts arise, it is advisable to check the analyser’s historical log to compare the period during which erratic readings took place against the known operating conditions. In addition, always check the analyser’s high and low measured values. Any discrepancies in the measured values may be symptomatic of a number of potential issues. These could include incorrect calibration or variations in ambient temperature and humidity conditions that could be affecting the sample. If a filter is fitted prior to the analyser, there is also the possibility that some important parameters may be being filtered out.
4. Check calibration
Ensuring that your analyser is properly calibrated will have a material impact on its ability to deliver continued measurement accuracy. Errors can arise when changing reagents, with potential failures arising if the reagents are not fitted properly, if tubing is blocked or restricted or if the reagent lines were not properly primed. Failure of a routine calibration can also be symptomatic of issues within the analyser, such as blocked tubes and valves, leaking seals or contamination of the reference standard.
5. Check the sensor
It is worth checking analyser systems that are in direct contact with the process, as sensors can become obstructed or damaged as they often bear the brunt of any adverse conditions. In waste water treatment processes in particular, sensors can quickly become fouled by the build-up of organic matter or subjected to abrasion by sediment. The accuracy of electrochemical sensors used in dissolved oxygen monitoring applications, for example, can quickly become impaired due to fouling of the sensor membrane.
6. Replace components regularly
Follow the manufacturer’s recommendations regarding the replacement of any key components. Depending on the type of analyser being used and its application, consider changing parts such as tubing, measurement cells and seals and diaphragms on a yearly or bi-annual basis.
7. Create a maintenance routine
Given the diverse range of parameters that are measured in water and waste treatment processes, it is difficult to set a universal rule when it comes to setting the frequency of maintenance checks. In most cases, it is possible to create a maintenance routine by monitoring the nature of past breakdowns with any issues that have been recorded, such as fouling, abrasion or measurement variations. By using these as a guideline, it will then be possible to create a maintenance routine that can help to safeguard against future occurrences. In addition, information from the device manufacturer should also be taken into account in order to further optimise performance.
8. Consider the lifecycle
Another factor that can affect the way that an analyser is maintained is where it sits in its lifecycle. This should take into account the model, age and make of the analyser. Models that are at the start of their product lifecycle will be well-supported, with ready spares availability. This may differ for models from older product ranges, where spares and the expertise to service them may not be as readily available. Dealing with reputable manufacturers can help to minimise this, as they are more likely to be able to support older products or else to recommend a suitable upgrade path with an alternative replacement.
-ABB offers comprehensive servicing and support for continuous water analysers. For more information, call 01480 488080 or email email@example.com.
- Pump maintenance: keeping it pumping Maintaining pumps in top condition is a vital consideration in the totex equation for pumping assets, and water utilities... Read More >
- Front Line: Simon Lee, Water Parks Ranger, Northumbrian Water WWT meets Simon Lee, one of Northumbrian Water's water park rangers who works at the company's fisheries and waterside... Read More >
- Getting to Grips with... Cryptosporidium All was quiet on the Cryptosporidium front but now the water industry's bête noire has re-surfaced. What can water... Read More >
- Opinion: A smarter way to read the water network Smart metering technology could be the key to closing water companies' knowledge gaps about usage in their network -... Read More >
- Smart thinking: The digital asset management revolution With many of the pieces now in place for a digital asset management revolution, AI and machine learning solutions are set... Read More >
- Networking skills: MUS helps Leeds to smarten up Morrison Utility Services has delivered a groundbreaking smart water network project in Leeds on behalf of Yorkshire Water Read More >
- Opinion: Protecting our water assets against cyber threats It's vital that engineers start to lead the conversation on cyber security for operational technology in the water... Read More >
- Innovation Zone: Waterfall out to make a splash Fitted near a building's stopcock and supplying cloud-based, accessible information about water use, an ‘intelligent water... Read More >